Buy more than* S$200* and get FREE SHIPPING
Due to the maintenance, we will close our website on June/19 9:00-10:00 and June/23 0:00-0:30 JST.
All customers cannot shop on our website during the time. Thank you.
23deg C+/-2deg C 45 to 75% or less no condensation
Current between Terminals
Current through the DUT (measured between terminals): measure resistance, including resistance range, Approximate values 0-80 ? A ohm x 1 k 0 To 800 ?m A ohm x 100 0 To 8 mA ohm x 10 0-80 MA ohm x 1
x1/x10/x100/x1k/x10kohm tolerance-scale length within +/- 3%
120m tolerance-maximum surfacing value within +/-4% internal resistance 4kohm/v 3 / 12 / 30 / 120 / 300 / 1000 V / 30KV tolerance - maximum surfacing value +/-2.5% less than internal resistance of 50kohmohm/v (1000V 15k range ohm/v)
Operating Temperature and Humidity Range
80% RH at 5degreeC to 31degreeC(max.) 31: At 40 degree c 80-50% RH decreasing linearly at 31 to 40degreeC There is no condensation
Condenser Capacity (microF)
C1 range - 50 to 0.2 tolerance. scale length +/- 6% C2 range - 0.01 to 20 tolerance. scale length +/- 6% C2 range -1 to 2,000 approximate values
Internal magnet type taut band(drop shock proof) 15 micro A
Polarity switch(switching + and - og measurement terminal)
DC Current Transfer Ratio
Transistor DC current gain hFE = lc/lb Approximate values 0 To 1000
0.5A/250V (5x20-mm ceramic pipe) Quick cutting model Breaking capacity 1500
30micro/0.3m/3m/30m/0.3A tolerance - maximum surfacing value within +/-2.5% l(30microA, 0.3 A-range is within +/-3%)
Altitudes below 2000m - pollution level 2 - indoor use
3 / 12 / 30 / 120 / 300 / 750 V tolerance-maximum surfacing value within +/-3% (Less than 12V within +/-4%) internal resistance of 8 k ohm/v
Outside Dimension (Height x Width x Depth) (mm)
R6 (AA) Type 1.5V (2), 6F22 (lamination type) 9V (1)
Test lead (TL-21) 1 set Spare fuse (product number F1176) 1 clip lead (CL-506) 1
Both red and black are covered up to the pointed front of the test lead tip, so it is difficult to short-circuit, but it is difficult to use it with the side of the bar etc. Because the movement of the needle is slow, feeling is hard to grasp when searching by hitting Jeong Chun.
I used an analog tester from before but it has broken and recently it has been increasing to use digital multimeter. But it seems easy to use analog tester familiar with it. I would like to use it for a long time.